Our TEC controllers need to be ordered with a hardware configuration chosen by the customer. The TEC controllers are configured and tested internally before shipping. This article is intended to clarify questions about hardware configuration.
- TEC Controllers Models
- Order Help / Options Assistant
- Sensor Options
- Terminal Options
We offer the following TEC Controllers:
-Typical cooling power
±0 – 1.2 A
0 – 9.6 V
±0 – 4 A
0 – 21 V
±0 – 10 A
0 – 21 V
±0 – 16 A
0 – 30 V
|1 Bipolar channel||TEC-1092||TEC-1091||TEC-1089||TEC-1090|
|2 Bipolar channels||TEC-1092||TEC-1122
|4 Unipolar channels
Order Help / Options Assistant
The following hardware configurations are available in addition to the models:
A few examples:
Model: TEC-1091 Sensor Type: PT100 Terminal Configuration: SCREW
Model: TEC-1161-4A Sensor Type: PT1000 Terminal Configuration: PIN
Model: TEC-1122-SV Sensor Typ: NTC Terminal Configuration
Sensor Options- ?s?
The following sensor types are available for our TEC controllers:
|Sensor Type||Temperature Measurement Range|
|PT100||-100 to +200 °C|
|PT1000||-100 to +200 °C|
|NTC||-55 to +131 °C assuming an NTC thermistor 10k@25 °C (B approx. 3900) is used|
The NTC hardware configuration allows the connected thermistor to be measured up to a maximum of 1M ohms. If the connected thermistor is bigger than 1M Ohm, 1M Ohm is measured. The NTC configuration corresponds to the previous NTC1M version.
To use our TEC Controller with thermocouples type K, you need a TCI-1181 in addition to the TEC Controller with a VIN1 object sensor type configuration. The TCI-1181 is a very small thermocouple readout circuit that amplifies and linearizes the thermocouple output.
The TCI-1181 has the following hardware configuration:
- Thermocouple temperature range type: K or KH
- Thermocouple Connector (X1): NC (solder eye) or MC (standard mini socket Farnell S/N 2810611)
- Board Connector (X4): NC (solder eye) or SCREW (see below)
|Thermocouple||Temperature Measurement Range|
|Typ K||-195 to +320 °C for TEC-1089, TEC-1090, TEC-1091 HW v1.8, TEC-1122, TEC-1123
-195 to +400 °C for TEC-1091 HW v3.15 , TEC-1092, TEC-1061
|Typ K (high temperature range)||5 to +540 °C for TEC-1089, TEC-1090, TEC-1091 HW v1.8, TEC-1122, TEC-1123
5 to +760 °C for TEC-1091 HW v3.15 , TEC-1092, TEC-1061
Hardware and Software configuration TCI-1181
Pin Configuration and Mechanical Data
Top view of the board
- The PCB can be mounted with the mounting holes (shown in blue, distances shown in picture) with M2 screws and spacers. The bottom of the board has no components.
- X1 Thermocouple connector (optional): Omega PCC-SMP-K Miniature Thermocouple Connector, mates with common thermocouples (Farnell P/N 2810611)
- X4 Device connector: screw header. Alternate connector available on request.
Pinout (pin 1 is at the bottom in the above picture):
Pin Pin Name Funktion Connect to TEC Controller Pin 4 TOIA Positive Supply IA 3 TOIB GND IB 2 TOUA Positive Output Terminal UA 1 TOUB Negative Output Terminal UB
- Direct wire soldering (compatible with connectors):
Pin Pin Name Funktion Connect to TEC Controller Pin Z3 TOIA Positive Supply IA Z4 TOIB GND IB Z5 TOUA Positive Output Terminal UA Z6 TOUB Negative Output Terminal UB
In the service software only the area "Chx Object Voltage to Temperature Conversion" (Tab: Advanced -> Tab: Temperture Conversion -> Approx. in the middle) must be set.
Terminal Options -?t?
The following terminal options are available and can only be specified for TEC-1091 and TEC-1161-xA:
|SCREW||TEC-1091: 2.54 mm
TEC-1161: 2.54 mm
|PIN||TEC-1091: 2.54 mm
TEC-1161: 2.54 mm